LNEYA's low temperature recirculating chiller can be used with supporting instruments (such as rotary evaporators, fermentation tanks, chemical reactors, freeze-drying equipment, biopharmaceutical reactors, etc.), with large cooling capacity and fast speed. The low-temperature circulating chiller can conduct low-temperature and constant-temperature experiments in the open cold tank inside the machine, or circulate the low-temperature liquid outside to cool or keep other equipment at low temperature. There are many precautions when using low-temperature recirculating chillers. In order to prevent malfunctions, LNEYA lists the following precautions for your reference:
1. The installation environment of the low-temperature recirculating chiller must be dry and ventilated, and there should be no obstacles around the equipment, otherwise it will affect the normal use of the equipment.
2. The liquid medium added inside the low-temperature recirculating chiller is particularly important. The liquid medium is selected according to different temperatures to add different mediums. The position of the liquid is also very important. The liquid level of the medium must be about 30 mm below the edge of the liquid tank, otherwise the heating device will be damaged after power on.
5. The low-temperature recirculating chiller must do a good job in daily maintenance work, and must ensure that the appearance of the equipment is dry and clean. Doing a good job in the maintenance of the equipment is very important to the service life of the equipment.
6. After using the low-temperature recirculating chiller, be sure to turn off the switch button of the equipment first, and then unplug the power supply. Please do not forcibly unplug the machine's power supply during the operation of the machine, as this will cause great damage to the machine itself.