Solution to oil spill of reactor cooling and heating circulator
When we use some reactor cooling and heating circulators, we will find that the system pressure fluctuates up and down as the temperature rises, and the oil tank is overflowing. So what are the reasons for these conditions in the cooling and heating circulators and how to solve them?
There are three reasons for the above situation: residual water in the pipeline or water in the heat-conducting oil component causes oil-water mixing, water vaporization at high temperature causes unstable system pressure, and oil spills in the fuel tank; the system pressure is normal at room temperature, and when the temperature increases When it rises (above 80 degrees), the system pressure gauge shakes up and down, and when there is a large amount of white smoke in the fuel tank or the heat transfer oil overflows, it can be judged that the oil and water are mixed.
Method of dehydrating boiling oil:
Remove the upper cover of the machine and open the fuel tank cover. Set the temperature to about 80 degrees first, and then combine the system pressure gauge. When the system pressure is stable, increase it by one or two degrees. If the system pressure drops again at this time, the temperature should be lowered again immediately. After the pressure stabilizes, increase it once or twice. In this way, the oil temperature is controlled at about 130 degrees under the condition of stable pressure (not more than 130 degrees, otherwise the exhaust solenoid valve from the system to the fuel tank will be closed, causing the system water vapor to fail to evaporate through the fuel tank, and the temperature will be too high. The heat transfer oil is oxidized too fast and affects its service life.) Let it run for 1-2 hours for dehydration and boiling. After the system pressure is completely stabilized, the temperature can be raised again. If there is no change in the system pressure midway, the fuel tank lid can be closed normal production. (Note: The temperature can be set directly next time the machine is turned on, no need to cook oil again.)
The expansion volume of the fuel tank is not enough, causing the oil to overflow. The system pressure did not change significantly with the increase in temperature, but the oil level of the oil tank liquid level indicator tube was continuously rising. After overflow, it can be judged that the oil tank is too small and the expansion volume is not enough. Solution: Replace the fuel tank with a larger size.
The filling of heat-conducting oil exceeds the normal level of the fuel tank, which leads to insufficient expansion space of the fuel tank and the heat-conducting oil overflows. Solution: Drain the excess heat transfer oil from the oil drain to the normal level.
February 6, 2021If you need equipment to distill CBD, you need to scrutinize your choices, as the equipment can vary in its features and design. When you use equipment that produces a CBD distillate, you want to seek out a system made by a manufacturer that provides proven and reliable products. You should also work with a business that can train you on the use of the equipment you buy.view
December 5, 2022On November 18, 2022, it was sponsored by the Jiangsu Provincial Association for Science and Technology, the Provincial Taiwan Affairs Office, the Provincial Department of Industry and Information Tec...view
January 13, 2023Recently, the Jiangsu Provincial Productivity Promotion Center officially released the evaluation results of the unicorn enterprises and gazelle enterprises in 2022 in Jiangsu and the high-tech zone, ...view
January 11, 2018Industrial Chiller：Industrial Chiller varies from dynamic temp. Special Chiller：Special Chiller consist of Multi-reactor chiller (TCU).Environmental Chamber：The test object is connected to a testing adapter. Industrial Freezer：Industrial freezer consist oview
July 4, 2018What are the characteristics and functions of ultra low freezer: 1. Ultra low freezer can prevent the precision of machine equipment from being influenced by the change of oil temperature in the proce...view
January 11, 2018Temperature ranges from - 5 ~ - 150 ℃, can meet different temperature; Adopt environmental protection refrigerants; Famous brand semi-closed piston compressor, semi-closed bipolar piston compressor, semi-hermetic screw compressor, main brand are BOCK.view
April 10, 2018Working temperature range from -25℃to 200℃, other temperature range need to set to the widest temperature range(-50℃to 250℃)
Centralized control of 2~20 reactors, each reactor's and jacket temperature can be set independently. One system control multipleview
April 10, 2018A set of cooling heat exchanger, heat exchanger, through proportional adjusting valve control cold heat into the heat exchanger, then through the uniform medium input to the reactor jacketed to do heat exchange and temp. Control, system has inside expansiview
February 28, 2019Temperature control range：-85 ° C ~ 200 ° CPower range：2.5kW~25kWTemperature control accuracy：±0.3°CModelTES-8525WTES-8555WTES-85A10WTES-85A15WTES-85A25WTemp. Range-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃Heating P...view