Although the refrigeration system and control circuit of ultra low temperature freezer in laboratory are complex, the common faults are mainly circulating system and circuit part.
Following are several common faults of ultra-low temperature refrigerators and their troubleshooting methods:
1. After power supply is switched on, the air switch of the main power supply is automatically disconnected.
Elimination method: Check whether the compressor of refrigeration system is damaged or not, and replace the parts if damaged. After replacing the fittings, turn on the machine to check whether it is normal. The fault is usually caused by unstable power supply, because the * stage starts first, so the damage is usually in stage. In this case, as long as the delay regulated power supply is installed, the fault will not occur again.
2. Stop working before the temperature in the box reaches the set temperature.
Exclusion: Although the temperature in the box has not reached the set temperature, the panel display temperature has reached the set temperature. This is due to the damage of the thermistor which detects the temperature and the need to replace the parts. If there is no original accessories, a set of temperature control system can be re-installed to eliminate the failure.
3. Poor refrigeration effect may be refrigerant leakage
4. If the ultra-low temperature refrigerator does not start in winter, it may be that the thermostat regulation value is too small.
We are a ultra low temperature freezer manufacturer. Welcome to choose our products.
The semiconductor temperature test system is LNEYA's new device for semiconductor chip testing. So what range does the semiconductor temperature test system apply? Users need to understand the specific application range to select the appro...View Details
In many current chemical experiments, the vacuum coating high and low temperature circulation system can be used, making the high and low temperature refrigeration heating system supporting the vacuum coating system, and the operation is m...View Details
At present, manufacturers of hot and cold rapid temperature change test chambers are sprouting like spring bamboo shoots, but there are several manufacturers that really grasp the quality. When purchasing a test box, the buyer must first l...View Details
Before producing the industrial chiller, the chiller manufacturer will consider the applicable industry, refrigeration capacity, structure, etc., and consider the actual use environment during design and assembly to optimize some functions...View Details