Precautions for the use of cooling and heating circulation device for lithium battery testing
As a professional experimental test auxiliary instrument, the cooling and heating circulation device for lithium battery testing is the circulating supply end of the cold source and the heat source, with a wide operating range. It is used in pharmaceutical, chemical, biological, etc., to provide heat and cold sources for reactors, tanks, etc., and can also be used for heating and cooling of other equipment. The temperature control range is wide, the heat transfer medium does not need to be replaced in the whole process, and the heat transfer medium consumption is low. The normal installation and use of the lithium battery test cooling and heating cycle device in the handling of problems will directly affect the performance and service life of the instrument.
In the high temperature experiment, if the temperature change does not reach the experimental temperature value, you can check the electrical system and check the problem one by one. If the temperature rises very slowly, check the air circulation system to see if the regulating baffle of the air circulation is open normally; otherwise, check if the motor operation of the air circulation is normal. If the temperature overshoot is severe, then you need to adjust the PID parameter settings. If the temperature rises directly and the maintenance is over-temperature, then the controller has a problem and the controller must be replaced.
The low temperature does not reach the goal of the experiment. Observe whether the temperature changes slowly, or whether the temperature has a rising trend after reaching a certain value. The former is to check whether the work room is dried before the low temperature experiment, and then the experiment samples are put into the work room after the experiment is done after the work room is kept dry. Are there too many experimental samples in the operation room, so that the wind in the operation room cannot circulate adequately? After removing the above reasons, it is necessary to consider whether it is a fault in the refrigeration system. In this way, it is necessary to ask the manufacturer's professional staff to perform maintenance. The latter phenomenon is caused by the poor operating environment of the equipment. The ambient temperature and location (the distance between the back of the box and the wall) of the equipment must meet the requirements (there are rules in the equipment operation instructions).
When there is a problem with the cooling and heating circulation device for the lithium battery test suddenly during the experimental operation, the corresponding fault display prompt and an audible alarm prompt will appear on the control instrument. The operator can quickly check which category it belongs to in the chapter on the fault removal in the operation of the equipment, and then ask professionals to quickly remove the fault to ensure the normal progress of the experiment. There will be other phenomena in the use of other environmental laboratory equipment, so specific phenomena must be analyzed and eliminated.
February 6, 2021If you need equipment to distill CBD, you need to scrutinize your choices, as the equipment can vary in its features and design. When you use equipment that produces a CBD distillate, you want to seek out a system made by a manufacturer that provides proven and reliable products. You should also work with a business that can train you on the use of the equipment you buy.view
January 11, 2018Industrial Chiller：Industrial Chiller varies from dynamic temp. Special Chiller：Special Chiller consist of Multi-reactor chiller (TCU).Environmental Chamber：The test object is connected to a testing adapter. Industrial Freezer：Industrial freezer consist oview
July 4, 2018What are the characteristics and functions of ultra low freezer: 1. Ultra low freezer can prevent the precision of machine equipment from being influenced by the change of oil temperature in the proce...view
July 4, 2018The refrigeration system of cold plate freezer adopts secondary super-cooler technology with the rapid refrigeration and low limiting temperature. It uses imported brand name semi-hermetic piston comp...view
January 11, 2018Temperature ranges from - 5 ~ - 150 ℃, can meet different temperature; Adopt environmental protection refrigerants; Famous brand semi-closed piston compressor, semi-closed bipolar piston compressor, semi-hermetic screw compressor, main brand are BOCK.view
April 10, 2018Working temperature range from -25℃to 200℃, other temperature range need to set to the widest temperature range(-50℃to 250℃)
Centralized control of 2~20 reactors, each reactor's and jacket temperature can be set independently. One system control multipleview
April 10, 2018A set of cooling heat exchanger, heat exchanger, through proportional adjusting valve control cold heat into the heat exchanger, then through the uniform medium input to the reactor jacketed to do heat exchange and temp. Control, system has inside expansiview
February 28, 2019Temperature control range：-85 ° C ~ 200 ° CPower range：2.5kW~25kWTemperature control accuracy：±0.3°CModelTES-8525WTES-8555WTES-85A10WTES-85A15WTES-85A25WTemp. Range-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃Heating P...view