1. Temperature selection
(1) The climatic zones of the region; Whether the product is exposed to the solar radiation environment near the heating device; Surrounding air, solar radiation and ways of heat transfer from heating devices to products; The actual technical state of the product (open or shaded).
(2)Use similar product temperature. Similar products refer to those with similar USES, performance and installation locations, such as upgrading products
2. The conditions of environmental chamber sales
(1) High-temperature storage
Thermostatic storage: after the sample reaches temperature stability (the temperature of internal components, components, modules, components, etc., also reaches stability) in the high-temperature environment, it can be maintained for at least 2 hours, and the longest can be 48 hours. If the storage time is over 48 hours, the temperature stability time will be included.
Cyclic storage: at least 7 cycles (24h for one cycle). If the key material or high temperature sensitive material is affected by high temperature environment, the cycle number should be increased appropriately.
(2) Hot operation in environmental chamber sales
Thermostatic work: after samples reached the stable in high temperature working environment(judgment methods: the differences of the internal heat of samples of the largest feature or key components should be less than 2 ℃ per hour , If the temperature of the internal element (part) cannot be measured, the temperature time shall be determined according to the thermal analysis to ensure that the temperature of the whole sample is stable)keep it at least 2 h, and then let the sample work, and make it on the specified working temperature to achieve stability, began to test the functionality and performance. After the test is completed, the sample will stop working and the air temperature in the box will be adjusted to the standard atmospheric conditions. After reaching the stability under the standard atmospheric conditions, the overall appearance, function and performance will be tested.
3. Temperature monitoring in environmental chambers
When monitoring the test temperature, the position of the temperature sensor should ensure that the influence of thermal diffusion is negligible, and pay attention to avoid the influence of thermal radiation.
The maintenance and maintenance of an ultra-low temperature refrigerator are very important for normal usage and prolonging its life. If the inaccurate temperature control often results in ...View Details
The cryogenic refrigeration system is different from air conditioning system, the medium and low temperature cryogenic refrigeration system in a common sense. It is more special in refri...View Details
At present, manufacturers of hot and cold rapid temperature change test chambers are sprouting like spring bamboo shoots, but there are several manufacturers that really grasp the quality. ...View Details
A bioreactor is a reaction system in which a naturally occurring microorganism or a microorganism having a special degrading ability is inoculated into a liquid phase or a solid phase. The ...View Details