The key points of using cryogenic refrigerator for cooling reaction
The cryogenic refrigerator is also called chiller or cryogenic refrigeration circulator. It is an experimental instrument that can achieve low temperature -110 degrees refrigeration. This kind of equipment is often used in single-fluid refrigeration tests. It is not the same type of equipment as general space refrigeration, cold storage, and refrigerator. So, what are the key points to pay attention to when using low-temperature refrigerators?
1. When the low-temperature refrigerator is used for the first time, use a multimeter to check whether the power plug connected to the device is normal, whether there is a short circuit or a bad state, if there is no abnormality, it can be used normally;
2. After the low-temperature refrigerator is connected to the power supply, add the corresponding circulating water according to the requirements, turn on the power switch on the equipment, and the circulating water starts to run;
3. Set the temperature range of the low-temperature refrigerator, set the high and low temperature ranges, and make observations. If the temperature range is large, continue to adjust the temperature range;
4. Low-temperature refrigeration circulating water is generally distilled water or manufacturer's circulating water. Do not use inlet water or river water, which will cause the circulation pipeline to block. If you use tap water, it is recommended to clean the pipeline every year;
5. Check the piping, refrigerant, etc. of the cryogenic refrigerator every six months for leakage and leakage, which can be distinguished according to the display parameters of the pressure gauge. If the data is far from the usual data, you can make a comparison;
6. The refrigeration system of low-temperature refrigerators needs to formulate an annual cleaning plan, including: heat dissipation pipes, condensing parts, cooling water towers, etc., to avoid system blockage and make the refrigeration system run more suitable. 7. If the low-temperature refrigerator is not used for a long time, it is recommended to turn off the water pump, compressor and power circuit switches of the equipment.
to sum up Low temperature refrigerators generally use pid temperature control technology. Different models of different manufacturers have different temperature accuracy. There are two types of slotting or fully enclosed. The cooling method also has two types of air cooling and water cooling. In order to achieve a better effect of reducing the temperature of the material, it is necessary Choose carefully.
February 6, 2021If you need equipment to distill CBD, you need to scrutinize your choices, as the equipment can vary in its features and design. When you use equipment that produces a CBD distillate, you want to seek out a system made by a manufacturer that provides proven and reliable products. You should also work with a business that can train you on the use of the equipment you buy.view
January 11, 2018Industrial Chiller：Industrial Chiller varies from dynamic temp. Special Chiller：Special Chiller consist of Multi-reactor chiller (TCU).Environmental Chamber：The test object is connected to a testing adapter. Industrial Freezer：Industrial freezer consist oview
July 4, 2018What are the characteristics and functions of ultra low freezer: 1. Ultra low freezer can prevent the precision of machine equipment from being influenced by the change of oil temperature in the proce...view
July 4, 2018The refrigeration system of cold plate freezer adopts secondary super-cooler technology with the rapid refrigeration and low limiting temperature. It uses imported brand name semi-hermetic piston comp...view
January 11, 2018Temperature ranges from - 5 ~ - 150 ℃, can meet different temperature; Adopt environmental protection refrigerants; Famous brand semi-closed piston compressor, semi-closed bipolar piston compressor, semi-hermetic screw compressor, main brand are BOCK.view
April 10, 2018Working temperature range from -25℃to 200℃, other temperature range need to set to the widest temperature range(-50℃to 250℃)
Centralized control of 2~20 reactors, each reactor's and jacket temperature can be set independently. One system control multipleview
April 10, 2018A set of cooling heat exchanger, heat exchanger, through proportional adjusting valve control cold heat into the heat exchanger, then through the uniform medium input to the reactor jacketed to do heat exchange and temp. Control, system has inside expansiview
February 28, 2019Temperature control range：-85 ° C ~ 200 ° CPower range：2.5kW~25kWTemperature control accuracy：±0.3°CModelTES-8525WTES-8555WTES-85A10WTES-85A15WTES-85A25WTemp. Range-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃Heating P...view