Ultra-low temperature freezer power supply and electronic control instrument system description
Ultra-low temperature freezer needs to be tested after a long time of use. What should be paid attention to when the power supply and electronic control instrument system are inspected?
Check the ultra-low temperature freezer power supply and the electronic control instrument system. Before starting the power, check whether the power distribution capacity matches the unit power, and whether the selected cable diameter can withstand the host operating current. Check whether the electrical system is consistent with the unit. The unit is electrically: three-phase five-wire system (three phase lines, one neutral line, one ground line, 380V ± 10%). Check if the power supply line of the compressor is connected tightly. If it is loose, retighten it. The torque at the compressor wiring is 500kg/cm. Due to factors such as long-distance transportation and lifting, the screws may be loose. Failure to do so may result in damage to electrical components (such as air switches, AC contactors, etc.) and compressors in the main control cabinet.
Use a multimeter to carefully check all the electrical circuits of the ultra-low temperature freezer, check that the wiring is properly installed in place; use megaohm measurement to make sure there is no short circuit in the case; check that the ground wire is properly installed in place, and the insulation resistance to ground is greater than 2MΩ; check if the power cable is suitable. Capacity requirements. Check if the disconnect switch is installed on the power supply line of the supply unit. All the wiring of the main circuit in the control cabinet and the external wiring of the control circuit can be energized after comprehensive inspection (such as crankcase oil heater, compressor electronic protector, circulating water temperature sensor, target flow switch wiring, pump connection) Control, etc.); Check if the terminal bolts are tightened and there is no looseness.
The performance of the power supply and the electronic control instrument are more important in the operation of the ultra-low temperature cold box, so regular inspection is recommended.
- December 17, 2020December 16-18
Booth number: W5P08
Shanghai New International Expo Center (Pudong)view
- February 6, 2021If you need equipment to distill CBD, you need to scrutinize your choices, as the equipment can vary in its features and design. When you use equipment that produces a CBD distillate, you want to seek out a system made by a manufacturer that provides proven and reliable products. You should also work with a business that can train you on the use of the equipment you buy.view
- January 11, 2018Industrial Chiller：Industrial Chiller varies from dynamic temp. Special Chiller：Special Chiller consist of Multi-reactor chiller (TCU).Environmental Chamber：The test object is connected to a testing adapter. Industrial Freezer：Industrial freezer consist oview
- July 4, 2018What are the characteristics and functions of ultra low freezer: 1. Ultra low freezer can prevent the precision of machine equipment from being influenced by the change of oil temperature in the proce...view
- July 4, 2018The refrigeration system of cold plate freezer adopts secondary super-cooler technology with the rapid refrigeration and low limiting temperature. It uses imported brand name semi-hermetic piston comp...view
- December 17, 2020LNEYA On November 3-5, meet at Chongqing International Expo Center [CIPM] Booth S3-07-2view
- January 11, 2018Temperature ranges from - 5 ~ - 150 ℃, can meet different temperature; Adopt environmental protection refrigerants; Famous brand semi-closed piston compressor, semi-closed bipolar piston compressor, semi-hermetic screw compressor, main brand are BOCK.view
- April 10, 2018Working temperature range from -25℃to 200℃, other temperature range need to set to the widest temperature range(-50℃to 250℃)
Centralized control of 2~20 reactors, each reactor's and jacket temperature can be set independently. One system control multipleview
- April 10, 2018A set of cooling heat exchanger, heat exchanger, through proportional adjusting valve control cold heat into the heat exchanger, then through the uniform medium input to the reactor jacketed to do heat exchange and temp. Control, system has inside expansiview
- April 10, 2018Low Temperature Cooling Chiller LT model: LT-5018N，LT-5040N，LT-5062N，LT-5090N，LT-50A1N，LT-50A2N; Medium Temperature Range: -45℃～30℃. ASET multifunctional controller, control outlet temperature of water tankview
- February 28, 2019-45℃~+200℃
- February 28, 2019Temperature control range：-85 ° C ~ 200 ° CPower range：2.5kW~25kWTemperature control accuracy：±0.3°CModelTES-8525WTES-8555WTES-85A10WTES-85A15WTES-85A25WTemp. Range-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃Heating P...view