Ultra-low temperature freezer power supply and electronic control instrument system description
Ultra-low temperature freezer needs to be tested after a long time of use. What should be paid attention to when the power supply and electronic control instrument system are inspected?
Check the ultra-low temperature freezer power supply and the electronic control instrument system. Before starting the power, check whether the power distribution capacity matches the unit power, and whether the selected cable diameter can withstand the host operating current. Check whether the electrical system is consistent with the unit. The unit is electrically: three-phase five-wire system (three phase lines, one neutral line, one ground line, 380V ± 10%). Check if the power supply line of the compressor is connected tightly. If it is loose, retighten it. The torque at the compressor wiring is 500kg/cm. Due to factors such as long-distance transportation and lifting, the screws may be loose. Failure to do so may result in damage to electrical components (such as air switches, AC contactors, etc.) and compressors in the main control cabinet.
Use a multimeter to carefully check all the electrical circuits of the ultra-low temperature freezer, check that the wiring is properly installed in place; use megaohm measurement to make sure there is no short circuit in the case; check that the ground wire is properly installed in place, and the insulation resistance to ground is greater than 2MΩ; check if the power cable is suitable. Capacity requirements. Check if the disconnect switch is installed on the power supply line of the supply unit. All the wiring of the main circuit in the control cabinet and the external wiring of the control circuit can be energized after comprehensive inspection (such as crankcase oil heater, compressor electronic protector, circulating water temperature sensor, target flow switch wiring, pump connection) Control, etc.); Check if the terminal bolts are tightened and there is no looseness.
The performance of the power supply and the electronic control instrument are more important in the operation of the ultra-low temperature cold box, so regular inspection is recommended.
- July 4, 2018The refrigeration system of cold plate freezer adopts secondary super-cooler technology with the rapid refrigeration and low limiting temperature. It uses imported brand name semi-hermetic piston comp...view
- July 4, 2018What are the characteristics and functions of ultra low freezer: 1. Ultra low freezer can prevent the precision of machine equipment from being influenced by the change of oil temperature in the proce...view
- July 4, 2018Guanya is a leading role in heating cooling circulator. It provides the customers with proposal and technology of controlling the cold and heat, temperature control system with stable property, extend...view
- July 4, 2018The price of heating cooling circulator is determined by raw materials, manufacturing overhead, technical cost, impacting on the final quotation.The heating cooling circulator is made of shell and int...view
- July 4, 2018With the continuous development of enterprises, as a matching equipment used in the lad, the heating cooling circulator which acts as a brand for manufacturers will be prosperous in the market, accele...view
- July 4, 2018The refrigeration principle of the so-called low temperature chiller is to use the artificial method, compressor refrigeration, through energy consumption, in a cycle process, in the process that cool...view
- January 11, 2018Temperature ranges from - 5 ~ - 150 ℃, can meet different temperature; Adopt environmental protection refrigerants; Famous brand semi-closed piston compressor, semi-closed bipolar piston compressor, semi-hermetic screw compressor, main brand are BOCK.view
- March 30, 2018Performance characteristics：1. In the refrigeration system, the condensation adopts air cooling mode, which has a wide temperature control range, which can meet the user's various needs.2. With a unique b...view
- April 8, 2018Performance characteristics: 1. With wide temperature control range, can meet the user's various needs.2. With a unique balanced temperature control method, with stable and balanced heating capacity, high precision and high stable temperature control canview
- April 10, 2018Working temperature range from -25℃to 200℃, other temperature range need to set to the widest temperature range(-50℃to 250℃)
Centralized control of 2~20 reactors, each reactor's and jacket temperature can be set independently. One system control multipleview
- April 10, 2018A set of cooling heat exchanger, heat exchanger, through proportional adjusting valve control cold heat into the heat exchanger, then through the uniform medium input to the reactor jacketed to do heat exchange and temp. Control, system has inside expansiview
- April 10, 2018ModelHQ-0540HQ-1040HQ-2540HQ-5040HQ-0575WHQ-1075WHQ-2575WHQ-5075WGas flow5m3/h10m3/h25m3/h50m3/h5m3/h10m3/h25m3/h50m3/hLowest Temp.-40℃-40℃-40℃-40℃-75℃-75℃-75℃-75℃Primary Temp.There is no-20℃...view