Reference method for using multiple reactor temperature control systems
The multi-reactor temperature control system is developed by LNEYA through years of trials using the industry's comparable refrigeration heating technology. When using multiple reactor temperature control systems, you should read the instruction manual for use.
A plurality of reactor temperature control systems are connected with a heat conductive synthetic oil and a power source, and a plurality of reactor temperature control system outlets are connected to the second site inlet, and the circulation device inlet port is connected to the second field outlet. Become a closed loop system. Multiple reactor temperature control systems use three-phase five-wire system (three live wires, one neutral wire, one ground wire) to connect the power supply. If the line continues to be in error (the phase protector indicator in the circuit is red) Replace any two live line positions until the phase changer indicator is green. Open the top cover of the filling port, add the heat transfer oil to the expansion oil level of the liquid storage tank (marked by the liquid level mirror), open and open the venting valve of the temperature control system of multiple reactors, and open the temperature control unit of multiple reactors. System power, press the liquid button, start adding heat transfer oil to the system until there is liquid overflow from the vent. Close the bleed valve, open the running button, set the temperature of the temperature control system of multiple reactors to 150 degrees, and continue to bleed and add liquid to the expansion position. Close the bleed valve, open the operation key of multiple reactor temperature control system, set the temperature to 25 degrees, continue to deflate and add liquid to the expansion position, and the test run is completed. The actual measured temperature is displayed when the microcomputer controller is powered up.
Parameter setting of multiple reactor temperature control system: Press SET button, the upper row shows SP lower row as the required setting value, press the up or down button to reach the desired setting value, and then press SET button to exit. Modify the internal parameters, press the SET button for more than 5 seconds, display the function menu, then click the SET button, press the LK to the password lock, and then press the up button to display the LK down button as 1. Click the SET button and press the desired parameter. Modify it in the same way. Press the SET button for more than 5 seconds. Turn on the power supply of multiple reactor temperature control systems, set the required temperature, and press the run key switch.
How to use the temperature control system of multiple reactors Everyone needs to take a closer look. LNEYA uses high-quality accessories, which is more reliable in operation.