Explosion-proof hot and cold machine winter maintenance instructions
When the explosion-proof type hot and cold integrated machine is used at the end of the day or when it is not used by the standby, be sure to disconnect the cooling water connection pipe, clean the remaining cooling water in the pipe, and store it in a dry place to prevent freezing and expansion. Damage to machine parts (such as coolers, pumps, connecting copper pipes). The method of removing residual water is as follows:
1. The water sequence series needs to disconnect the cooling water and hot water inlet and outlet pipes and open the corresponding ball valve. Take the air gun to the hot water (outlet) inlet and let the residual medium inside the system discharge from the cooling water inlet (cooling) The water outlet is discharged from the inlet due to the electromagnetic valve. It is determined that there is no water in the pipeline and it can be stopped.
2. The standard explosion-proof type hot and cold integrated machine needs to disconnect the connecting pipe of the cooling water inlet and outlet, open the ball valve and let the machine be in a cooling state. When the cooling electromagnetic valve is opened, take the air gun and blow it into the cooling water inlet to make it from the water outlet. Discharge, determine that there is no water in the pipeline and then stop.
Kindly note, please be sure to disconnect the cooling water connection pipe when the equipment is not in use, and clean the remaining cooling water/heat transfer oil in the pipe, and store it in a dry place.
In the winter, when you do not need the explosion-proof type hot and cold machine, you need to pay attention to turn off the explosion-proof hot and cold machine and clean it as well, so as to avoid the residue affecting the performance of the explosion-proof hot and cold machine.
- July 4, 2018The refrigeration system of cold plate freezer adopts secondary super-cooler technology with the rapid refrigeration and low limiting temperature. It uses imported brand name semi-hermetic piston comp...view
- July 4, 2018What are the characteristics and functions of ultra low freezer: 1. Ultra low freezer can prevent the precision of machine equipment from being influenced by the change of oil temperature in the proce...view
- July 4, 2018Guanya is a leading role in heating cooling circulator. It provides the customers with proposal and technology of controlling the cold and heat, temperature control system with stable property, extend...view
- July 4, 2018The price of heating cooling circulator is determined by raw materials, manufacturing overhead, technical cost, impacting on the final quotation.The heating cooling circulator is made of shell and int...view
- July 4, 2018With the continuous development of enterprises, as a matching equipment used in the lad, the heating cooling circulator which acts as a brand for manufacturers will be prosperous in the market, accele...view
- July 4, 2018The refrigeration principle of the so-called low temperature chiller is to use the artificial method, compressor refrigeration, through energy consumption, in a cycle process, in the process that cool...view
- January 11, 2018Temperature ranges from - 5 ~ - 150 ℃, can meet different temperature; Adopt environmental protection refrigerants; Famous brand semi-closed piston compressor, semi-closed bipolar piston compressor, semi-hermetic screw compressor, main brand are BOCK.view
- April 10, 2018Working temperature range from -25℃to 200℃, other temperature range need to set to the widest temperature range(-50℃to 250℃)
Centralized control of 2~20 reactors, each reactor's and jacket temperature can be set independently. One system control multipleview
- April 10, 2018A set of cooling heat exchanger, heat exchanger, through proportional adjusting valve control cold heat into the heat exchanger, then through the uniform medium input to the reactor jacketed to do heat exchange and temp. Control, system has inside expansiview
- April 10, 2018Low Temperature Cooling Chiller LT model: LT-5018N，LT-5040N，LT-5062N，LT-5090N，LT-50A1N，LT-50A2N; Medium Temperature Range: -45℃～30℃. ASET multifunctional controller, control outlet temperature of water tankview
- February 28, 2019-45℃~+200℃
- February 28, 2019Temperature control range：-85 ° C ~ 200 ° CPower range：2.5kW~25kWTemperature control accuracy：±0.3°CModelTES-8525WTES-8555WTES-85A10WTES-85A15WTES-85A25WTemp. Range-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃-85℃～200℃Heating P...view